BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||16 April 2008|
|PDF File Size:||7.23 Mb|
|ePub File Size:||9.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Discharge sample a Place the pipette tip against the wall of the vessel. To print the manual completely, please, download it. Ask your supplier or manufac- turer for the form. Ensure that the tip is securely seated. For further information, rtansferpette to the accuracy table on pages 37 and Inspect the piston and seal for contamination. Hold the instrument vertically, with the sample in the tip for approximately 10 s. Checking The Volume Checking the Volume Depending on use, we recommend inspection manuao the instrument every 3 to 12 months.
Pipetting Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Comments to this Manuals Your Name. Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
The form can also be downloaded from www.
Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Inspect the piston and O-Ring-seal for contamination. Laboratory Equipment Brand dispensette Operating Manual pages. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment. Enter text from picture: Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions.
Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Optimum analysis results can only be obtained with quality tips. Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure. Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous nanual with medium density and low to medium brznd.
Autoclave the complete mqnual without any further disassembling.
BRAND Transferpette S Operating Manual
Got it, continue to print. Inspect the pipette tip cone for brajd. Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
It is beyond the scope of this manual to address all of the potential safety risks associated with its use in such applications. Eject the pipette tip. Don’t have an account?
Pipette shaft Tip cone Fig. Page 8 Pipetting c Let the pipetting button slide back slowly.
Limitations of Use The instrument is intended for the pipetting of liquids within the following limitations: The cycle can, however, be adjusted to individual requirements. If the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, maual can be made using the Easy Calibration Technique.
By pressing ‘print’ button you will print only current page. To do this, affix a tip, trandferpette aspirate a sample. Page of 25 Go. Safety Instructions Please read the following carefully!
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Just send in the instruments to be calibrated, accompanied by an indication of which kind of calibration you wish. Pipette tips are disposables items! Table of Contents Add to my manuals Add. Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of bramd as well as the failure to transferpftte the instructions of the operating manual.
Consult the manufacturer for use in temperatures outside of this range. Fitting the tip Use the correct tip according to the volume range or the color code. ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process. Parts will differ slightly depending on nominal volume of instrument. Don’t lay the instrument horizontal when the tip is filled.
Page 9 Transfeerpette Ejecting the tip Hold the pipette shaft over a suitable disposal container manuwl press the tip ejection key to the stop. Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity.
In order for the liquid to reach its end position, leave the tip immersed for a few seconds. Safety Instructions Safety Instructions Please read the following carefully!
Pipetting c Let the pipetting button slide back slowly.